Laboratoire d’Etudes en Géophysique et Océanographie Spatiales

Personal tools

This is SunRain Plone Theme

You are here: Home / Measurement Technique

# Measurement Technique

While ocean satellite observing systems generate comprehensive views of the ocean surface, sub-surface measurements are limited to moored instruments such as current meters, robotic instruments that roam the ocean drifting with the currents and surfacing at programmed times to download their data to satellites, research vessels with specialised deep ocean measuring systems, or expendable bathythermographs which are ocean temperature probes called 'XBT's'.

#### XBT's

Shaped like miniature torpedos, these probes contain a spool of fine copper wire which spins out of the XBT as it descends, and a thermister which is sensitive to temperature. Resistance across the thermister changes with temperature and this resistance is measured on board the ship as long as the copper wire remains intact.
Depth is inferred from time: we know how fast the XBT falls through the water so we can calculate how deep the instrument is at any point in its fall if we measure the time since it was dropped. Temperature recording stops when the XBT runs out of wire which breaks, terminating communication.

The launcher remains on board the ship. It also contains a spool of copper wire so the ship need not slow while the XBT is deployed - wire simply flows off of the ship-board spool as the ship moves, eliminating tension on the wires. The instrument is "expendable", meaning it cannot be retrieved after it is launched.

The launcher is linked to a computer system on the ship that records the resistance of the wire, calculates temperature from this resistance and depth from the time since launch, and saves this information to disk (as well as position, date and time of the launch) for processing after the ship returns to port. We regularly get profiles as deep as 800 m from this system.
The instruments cost about A\$140 each (2004) but their data is priceless.

#### Thermosalinograph

A second instrument, the thermosalinograph, measures surface temperature and salinity continuously from an intake pipe on the ship's hull. The continuous records are averaged every minute, providing a fine scale trace of the ocean's surface structure. The logging system for the thermosalinograph is linked to a GPS recorder and also records the ships meteorological data.
The Logging System records:
- Latitude and longitude
- Surface temperature and salinity
- Wind speed
- Barometric pressure

Starting in 2006, the thermosalinograph data will be transmitted in near real time.
These data are incorporated in the ORE-SSS at LEGOS.

Two computers on the bridge (controls XBT launch and logs the thermosalinograph data)

Document Actions